Laminar flow sampling car | clean sampling car
The sampling vehicle is also called clean sampling vehicle or laminar flow sampling vehicle. It is mainly used for sampling of raw and auxiliary materials except for pharmaceutical preparations and sterile preparations. The sampling vehicle is a vertical flow local air purification device.
Clean sampling vehicle USES:
Clean sampling car is widely used in medicine, pharmaceutical, electronic, precision instruments, instruments, food and other industries of special occasions of raw materials sampling.
Features of our sampling car/laminar flow sampling car:
The sampler produced by our company has the advantages of simple structure, beautiful appearance, flexible movement, economical benefit and convenient use, etc. It is the necessary equipment to carry out the sampler of article 26 of GMP in the pharmaceutical industry. The electric components used at the same time conform to gb1497-85 (electric safety standard); Meet the requirements of gb-191 (packaging standard). The equipment has a cleanliness level of 100 and can conveniently sample the raw and auxiliary materials of pharmaceutical preparations (except sterile preparations).
Clean sampling car electrical control panel for the computer light contact control, fan stepless speed regulation, arbitrary regulation of air volume, clean sampling car equipped with ultraviolet sterilization lamp, other specifications can be customized according to customer requirements or both ends of the installation of soft curtain.
Clean sampling car usage:
The clean sampling car shall be moved to the place where sampling is needed, and the sampling car shall be fixed (brake); Switch on the power, turn on the power switch, turn on the fan switch;
After the fan starts working, adjust the voltage to appropriate (generally 220V), and put it into the sampling bucket after the clean sampling car runs for 15 minutes;
After the sampling, remove the sampling bucket, turn off the fan and power supply, unplug the power cord, and move the sampling car to the original position.
Environmental requirements for clean sampling vehicles:
Normal working conditions: temperature 5 ℃ to 40 ℃
Relative humidity: 80%
Atmospheric pressure: 86 ~ 106KPa
Appearance: the cabinet is made of stainless steel or steel. The curtain is smooth and vertical without damage.
Foot wheel: flexible and reliable.
Illumination: the average illumination shall not be less than 300LX.
Average wind speed: 0.4 20%(adjustable)
Noise: no more than 65db (A)
Vibration: within the range of wind speed standard, the amplitude of sampling vehicle frame in X, Y and Z directions is less than or equal to 3 m.
General configuration of clean sampling vehicle:
1. Aluminum frame without baffle high efficiency filter filter efficiency: firstname.lastname@example.org m and G4 class initial effect filter, ensure the purification level, static 100 U.S. federal standard 209E;
2. Stainless steel baffles are arranged on both sides of the work area, and strip-shaped anti-static and dustproof curtain is placed in front and back to achieve the ideal dustproof effect;
3. Equipped with large air volume and low noise fan to ensure the wind speed is greater than 0.45m/s;
4. With the pressure difference table of imported brands, it can clearly indicate the pressure difference on both sides of the hepa filter and timely remind you to replace the hepa filter;
5. Adjustable speed fan design, stable speed performance, can ensure the same air volume under the state of the final resistance of the high efficiency filter;
6. With adjustable foot and universal wheel, it is light and convenient to move and position, stable and safe;
7. Stainless steel inner bottom hole plate can be easily placed and removed.